I had just finished refilling the nitrogen and powering up the NMR after the holiday shutdown and went through the normal startup routine (cf, ii, cfbsms). When I ran lockdisp I could not get the lockdisplay window to appear, appearing to be stuck on 'checking rs-channel' and failing after 11 attempts.
Any help in the matter would be appreciated.
System details: XwinNMR 2.7 on SGI O2 with IRIX. Bruker Avance DPX400.
Things checked already:
asked Jan 02 '13 at 06:37
Maybe it's just the problem of the software if other commands are OK.
answered Jan 07 '13 at 16:11